SPEAG has released DASY8 Module HAC2019. The new module provides a guided workflow that demonstrates the compliance of any wireless device with the latest Hearing Aid Compatibility standard ANSI C63.19-2019 and the latest Federal Communication Commission requirements. Module HAC2019 is also available for DASY6.
Demonstrating Hearing Aid Compatibility (HAC) is important for warranting the electromagnetic and operational compatibility of hearing aids and wireless devices. In February 2021, the Federal Communication Commission (FCC) announced the adoption of the 2019 edition of ANSI C63.19, the “American National Standard Methods of Measurement Compatibility Between Wireless Communications Devices and Hearing Aids,” which became effective June 4, 2021. The FCC is giving manufacturers a two-year transition period (i.e., until June 4, 2023) to adopt the requirements of the updated ANSI C63.19-2011 standard. It has also extended the volume control deadline to match this transition period. Beginning June 5, 2023, only test reports that comply with ANSI C63.2019 will be accepted.
Following SPEAG’s tradition of staying ahead of requirements, we started early with the development of Module HAC2019 for DASY8/6 users. This is a completely new software solution based on SPEAG’s latest graphical user interface technology that guides users through the compliance workflow by using specific Jupyter Notebooks. The DASY8/6 system is interfaced via a dedicated application programming interface.
During development, we have been meticulous in ensuring that we reuse as much as possible of the previous HAC hardware in order to save resources and costs for our customers. The only element that has been replaced is the audio interference analyzer (AIA), which has been superseded by the modulation and audio interference analyzer (MAIA) to evaluate the modulation interference factor (MIF).
DASY8/6 Module HAC2019 consists of four Jupyter Notebooks that represent the different steps of the compliance workflow:
MIF measurements with MAIA
Simplified verification of the radiofrequency (RF) test setup using RF emission calibration dipoles
Evaluation of the wireless device RF interference potential by determining the RF audio interference level (RFail)
Measurement of the baseband (audio frequency) magnetic T-Coil (Tele-Coil) signal from a wireless device.
The workflow can easily be implemented, as shown in this video:
MIF Measurements with MAIA
The MAIA allows verification of the MIF values used to calculate the RFail. The MAIA can be operated over the air interface using the built-in ultra-broadband planar log spiral antenna (supported frequency range: 698–6000 MHz) or in conducted mode using the coaxial SMA 50 Ω connector (supported frequency range: 300–6000 MHz).
Figure 1: MIF Measurement of a 5G new radio frequency range 1 frequency division duplex signal
RF Test Setup Verification
A dedicated Notebook has been developed for verifying the RF interference measurement setup. The user can select and optimize the measurement configuration in a list of predefined templates. Peak values are automatically detected for easy comparison with the measurement targets.
Figure 2: Interpolated E-field Distribution of a CD2450V3 Dipole
RF Audio Interface Level Determination
In Module HAC2019, the RFail is assessed using the (preferred) indirect test procedure described in ANSI C63.19-2019. A 50x50 mm region centered on the speaker location is scanned and the root mean square (RMS) electric field values are recorded at each point. Once the scan is completed, the RFail is obtained by adding the MIF value to the average steady RMS field strength averaged over the measurement area.
Figure 3: Interpolated E-field distribution of a wireless device after MIF scaling
Baseband Magnetic T-Coil Signal
The Jupyter Notebook developed for baseband magnetic T-Coil testing guides the users through the measurement steps described in the ANSI C63.19-2019:
Validation of the measurement equipment using a telephone magnetic field simulator (TMFS) or a Helmholtz coil
Determination of the wireless device drive levels for each audio signal
Measurement of the desired magnetic field with a voice-like signal (desired audio band magnetic (ABM) signal)
Measurement of the frequency response at the maximum location of the desired ABM signal
Measurement of the undesired broadband audio magnetic signal (undesired ABM signal)
The T-Coil coupling qualifying strengths, including determination of the primary and secondary groups, as well the magnetic field frequency response are computed by the post-processor and are available for visualization and reporting.
Figure 4: Evaluation of Primary and Secondary Groups according to ANSI C63.19-2019
The software installer will be available on November 15th.
For further information, please contact us at email@example.com or any of our sales channels.
SPEAG introduces a new series of tissue simulating liquids that are broad-band based on the environmental friendlier Tween ingredient and fully compliant with the new IEC/IEEE standards allowing ±10% tolerance on the liquid parameters.
The National Research Council Institute for Biodiagnostics (NRC-IBD) has teamed up with SPEAG to offer a unique software platform for magnetic resonance imaging (MRI) scanner design. The new platform seamlessly integrates powerful new tools for radiofrequency (RF) array design and post-processing and is the first software of its kind to combine all the tools needed to design RF-phased arrays into a single user friendly program.
This new product PRE - 10 is a 3 channel voltage divider which extends the operation of existing SPEAG probes for use in very high fields (> 5 kV/m; > 10 kW/kg). It is compatible with all our E-Field, H-field and SAR probes.
The SEMCAD X Team is pleased to announce the release of SEMCAD X V14.2 Aletsch, a major upgrade six months of intense deincorporatingvelopment work. This release offers a range of powerful new features, more robust handling, as well as new customization options.
The relevant SEMCAD X for Win XP/ Win 7 and 32/64bit platforms are available for download.
The new EASY4 software release V4.6 extends the usage to fields with large peak-to-average (PAR) ratios by adaption to the improved probe modulation signal calibration and also includes setting of Access Levels. The changes to the user interface are minimal and old setup and calibration files are still compatible with this version.
A new iSAR2 software release is now available. This release is the result of the popularity of our ultra-fast SAR scanners (iSAR Head and iSAR Flat) and great suggestions from our customers. Version 2.1 of the iSAR software has several new features, including a new multi-iSAR interface, a smart Combiner feature, and full extension of the software for frequencies up to 6 GHz. It also includes enhancements to our existing features, such as an improved Python interface and enhanced control of the CMU200.