Picture above shows DASY8 Module WPT V2.x
DASY8 Module WPT V2.4+ is based on the MAGPy Version 2 technology integrated in our DASY8 product line for high-precision robot-based evaluations of wireless power transfer (WPT) devices. It is the only system for fully automated compliance testing according to all international standards and national regulations, e.g., IEC PAS 63184, FCC KDB 680106 D01, ISED Canada SPR-002. While DASY8 Module WPT V2.4+ is the most accurate system for the evaluation of WPT devices and systems in the lab, the handheld MAGPy Version 2 system is designed for in situ evaluations of large WPT devices, such as e.g., vehicular charging systems, that cannot be brought to a lab.
DASY8 Module WPT is composed of the isotropic probe MAGPy-8H3D+E3D Version 2, the reference probe (MAGPy-RAφ), and the data acquisition system (MAGPy-DAS) mounted to the DASY8 robot via the emergency stop (MAGPy-ES). It measures the incident electric (E-) and magnetic (H-) fields in a volume from the surface of the device under test (DUT) using advanced field reconstructions to obtain a high-resolution (mm range) field distribution. The induced E-field distributions and specific absorption rate (SAR) are assessed with Sim4Life’s Quasi-Static EM Solver (P-EM-QS) using only the measured data. At each probe location, eight sets of isotropic H-field values and one set of isotropic E-field values are acquired in parallel. The dedicated graphical user interface (GUI) fully automates the testing workflow.
|Laboratory evaluation of WPT systems and any other local electromagnetic source not requiring magnetic (H-) field volume scans exceeding 2000 mm × 1000 mm × 1500 mm:
|Typical Workflows for Compliance Testing
Details about testing compliance according to specific regulations are provided in the following Application Notes:
The basic components of DASY8 Module WPT are:
The MAGPy-DAS includes:
The MAGPy-8H3D+E3D probe consists of eight isotropic H-field sensors and one isotropic E-field sensor:
Specifications of the MAGPy-RAφ reference amplitude and phase probe:
The following WPT verification sources are available:
|MAGPy-8H3D+E3D probe calibrated according to IEC PAS 63184 (ISO/IEC 17025 accredited)
|See chapters 8.4 – 8.7 in the manual of DASY8 Module WPT
Option: Advanced Sim4Life Evaluation Upgrade Package
DASY8 Module WPT is complete and allows compliance testing according to all international standards and national regulations. The Sim4Life upgrade package includes all software components required to go beyond and further assess device-specific user scenarios. The package includes:
(i) Sim4Life Framework
(ii) Upgrade to P-EM-QS Suite (Unstructured Solver)
(iii) MPI-QS Token (package of six)
(iv) Virtual Population Set for Complete Population Coverage (Whole Body and Hand Phantoms)
Posable Ella V3.x Model and Hand
(v) P-EM-QS Link for Complete ViP Package
DASY8 Module WPT is fully compliant with:
DASY8 Module WPT V2.4+ is compatible with:
Updated Application Notes are available to guide testing in compliance with the different regulations. If you have questions or have received specific requests from the regulators, please send an email to firstname.lastname@example.org and we will respond within 24 hours.
Note that SPEAG's products are always fully compatible with the latest standards, and national regulations. We are members of the IEC/IEEE/ANSI standardization groups and are in continuous contact with the regulators to achieve full compatibility even prior to the release of any standards and new regulations.
|DASY6 Module WPT
|See product lifetime
February 15, 2024 (DASY8 Module WPT V2.4+)