After the successful launch of the first version of our hearing aid compatibility module, DASY8/6 Module HAC V1.0, in November 2021, SPEAG has further improved the features and workflow of the module based on valuable customer feedback. Streamlined procedures and improved visualizations now make it easier and smoother to demonstrate compliance with ANSI C63.19-2019.
Module HAC V1.2 supports the latest hardware revision of the Audio Magnetic Measurement Instrument (AMMI). Based on the Roland Rubix22 USB audio interface, it is fully compatible with Windows 10/11.
In Module HAC V1.2, the hearing aid compatibility test arch position and probe calibration results are persistent. It is no longer necessary to import them after restarting the Jupyter notebook – the last used values are automatically restored.
The audio signal emitted by the wireless device under test may be delayed due to the codec latency. In Module HAC V1.2, this parameter no longer needs to be evaluated manually: it is now determined automatically in post-processing.
The field viewer comes with many new features: for example, users can now edit the color bar range and export all displayed data. In addition, the spectrally weighted undesired magnetic noise can now be visualized.
The communication system selection has also been further improved: the layout is more user-friendly and the most frequently used test channels (low/middle/high) are more prominently highlighted.
The post-processing algorithm for evaluating the undesired magnetic field in the audio band has been improved to avoid overestimation of when low-frequency components dominate.
DASY8/6 Module HAC V1.2, together with the updated manuals, can be downloaded here.
Your feedback helps us to constantly improve our products and services. If you have thoughts and suggestions, please reach out to us at email@example.com.
For further information on DASY8/6 Module HAC V1.2 or any of our other products, please contact us at firstname.lastname@example.org or via any of our sales channels.