SPEAG has released DASY8 Module HAC2019. The new module provides a guided workflow that demonstrates the compliance of any wireless device with the latest Hearing Aid Compatibility standard ANSI C63.19-2019 and the latest Federal Communication Commission requirements. Module HAC2019 is also available for DASY6.
Demonstrating Hearing Aid Compatibility (HAC) is important for warranting the electromagnetic and operational compatibility of hearing aids and wireless devices. In February 2021, the Federal Communication Commission (FCC) announced the adoption of the 2019 edition of ANSI C63.19, the “American National Standard Methods of Measurement Compatibility Between Wireless Communications Devices and Hearing Aids,” which became effective June 4, 2021. The FCC is giving manufacturers a two-year transition period (i.e., until June 4, 2023) to adopt the requirements of the updated ANSI C63.19-2011 standard. It has also extended the volume control deadline to match this transition period. Beginning June 5, 2023, only test reports that comply with ANSI C63.2019 will be accepted.
Following SPEAG’s tradition of staying ahead of requirements, we started early with the development of Module HAC2019 for DASY8/6 users. This is a completely new software solution based on SPEAG’s latest graphical user interface technology that guides users through the compliance workflow by using specific Jupyter Notebooks. The DASY8/6 system is interfaced via a dedicated application programming interface.
During development, we have been meticulous in ensuring that we reuse as much as possible of the previous HAC hardware in order to save resources and costs for our customers. The only element that has been replaced is the audio interference analyzer (AIA), which has been superseded by the modulation and audio interference analyzer (MAIA) to evaluate the modulation interference factor (MIF).
DASY8/6 Module HAC2019 consists of four Jupyter Notebooks that represent the different steps of the compliance workflow:
MIF measurements with MAIA
Simplified verification of the radiofrequency (RF) test setup using RF emission calibration dipoles
Evaluation of the wireless device RF interference potential by determining the RF audio interference level (RFail)
Measurement of the baseband (audio frequency) magnetic T-Coil (Tele-Coil) signal from a wireless device.
The workflow can easily be implemented, as shown in this video:
MIF Measurements with MAIA
The MAIA allows verification of the MIF values used to calculate the RFail. The MAIA can be operated over the air interface using the built-in ultra-broadband planar log spiral antenna (supported frequency range: 698–6000 MHz) or in conducted mode using the coaxial SMA 50 Ω connector (supported frequency range: 300–6000 MHz).
Figure 1: MIF Measurement of a 5G new radio frequency range 1 frequency division duplex signal
RF Test Setup Verification
A dedicated Notebook has been developed for verifying the RF interference measurement setup. The user can select and optimize the measurement configuration in a list of predefined templates. Peak values are automatically detected for easy comparison with the measurement targets.
Figure 2: Interpolated E-field Distribution of a CD2450V3 Dipole
RF Audio Interface Level Determination
In Module HAC2019, the RFail is assessed using the (preferred) indirect test procedure described in ANSI C63.19-2019. A 50x50 mm region centered on the speaker location is scanned and the root mean square (RMS) electric field values are recorded at each point. Once the scan is completed, the RFail is obtained by adding the MIF value to the average steady RMS field strength averaged over the measurement area.
Figure 3: Interpolated E-field distribution of a wireless device after MIF scaling
Baseband Magnetic T-Coil Signal
The Jupyter Notebook developed for baseband magnetic T-Coil testing guides the users through the measurement steps described in the ANSI C63.19-2019:
Validation of the measurement equipment using a telephone magnetic field simulator (TMFS) or a Helmholtz coil
Determination of the wireless device drive levels for each audio signal
Measurement of the desired magnetic field with a voice-like signal (desired audio band magnetic (ABM) signal)
Measurement of the frequency response at the maximum location of the desired ABM signal
Measurement of the undesired broadband audio magnetic signal (undesired ABM signal)
The T-Coil coupling qualifying strengths, including determination of the primary and secondary groups, as well the magnetic field frequency response are computed by the post-processor and are available for visualization and reporting.
Figure 4: Evaluation of Primary and Secondary Groups according to ANSI C63.19-2019
The software installer will be available on November 15th.
For further information, please contact us at firstname.lastname@example.org or any of our sales channels.
SPEAG is pleased to introduce the next generation in fast, high-precision Compliance SAR (cSAR3D) Systems, a system concept that had been developed during the last months to meet the various requests/suggestions of our highly valued customers.
We are pleased to announce that the frequency range of DAK has been extended. A novel algorithm has been implemented that extends the dielectric measurement capability of the DAK-12 Probe to 10 MHz - 5 GHz.
The SEMCAD X Team is pleased to announce the release of SEMCAD X V14.8.0 Aletsch, a major upgrade offering a range of powerful new Modules and features, more robust handling as well as new customization options.
We are proud to announce a breakthrough in EM probe systems. The Time-Domain-Sensor (TDS) is the basis of our novel product line of high-performance active optical radio-frequency field sensor systems. The TDS concept allows the measurement of the full complex vectorial EM field distribution (amplitude and phase) in the time and frequency domains. First TDS probes for E- and H-field measurements in the 10MHz to 6GHz frequency range will become available in the next quarter.
Our first DAK shipments were made to customers in December. We are very pleased about the successful launch and strong customer interest in this new product line for high-precision dielectric measurements. Expansion of the frequency range and applications for the DAK product line will continue in 2012.
SPEAG and Dymstec are delighted to announce the opening of the joint SPEAG Calibration Laboratory Korea, Inc. The laboratory is now fully operational and completed its first iSAR calibrations this month. iSAR calibration is performed using the same high-quality equipment and procedures as the SPEAG iSAR calibration laboratory in Zurich, with the additional benefit of much faster iSAR calibration service for many Asian customers.
DAK is our new product line for high-precision dielectric measurements. Our probes are optimized for precision, robustness and ease of use. The software is flexible and operates with the most popular network analyzers on the market. The DAK system is calibrated by our ISO/IEC 17025 accredited laboratory.