SPEAG has released DASY8 Module HAC2019. The new module provides a guided workflow that demonstrates the compliance of any wireless device with the latest Hearing Aid Compatibility standard ANSI C63.19-2019 and the latest Federal Communication Commission requirements. Module HAC2019 is also available for DASY6.
Demonstrating Hearing Aid Compatibility (HAC) is important for warranting the electromagnetic and operational compatibility of hearing aids and wireless devices. In February 2021, the Federal Communication Commission (FCC) announced the adoption of the 2019 edition of ANSI C63.19, the “American National Standard Methods of Measurement Compatibility Between Wireless Communications Devices and Hearing Aids,” which became effective June 4, 2021. The FCC is giving manufacturers a two-year transition period (i.e., until June 4, 2023) to adopt the requirements of the updated ANSI C63.19-2011 standard. It has also extended the volume control deadline to match this transition period. Beginning June 5, 2023, only test reports that comply with ANSI C63.2019 will be accepted.
Following SPEAG’s tradition of staying ahead of requirements, we started early with the development of Module HAC2019 for DASY8/6 users. This is a completely new software solution based on SPEAG’s latest graphical user interface technology that guides users through the compliance workflow by using specific Jupyter Notebooks. The DASY8/6 system is interfaced via a dedicated application programming interface.
During development, we have been meticulous in ensuring that we reuse as much as possible of the previous HAC hardware in order to save resources and costs for our customers. The only element that has been replaced is the audio interference analyzer (AIA), which has been superseded by the modulation and audio interference analyzer (MAIA) to evaluate the modulation interference factor (MIF).
DASY8/6 Module HAC2019 consists of four Jupyter Notebooks that represent the different steps of the compliance workflow:
MIF measurements with MAIA
Simplified verification of the radiofrequency (RF) test setup using RF emission calibration dipoles
Evaluation of the wireless device RF interference potential by determining the RF audio interference level (RFail)
Measurement of the baseband (audio frequency) magnetic T-Coil (Tele-Coil) signal from a wireless device.
The workflow can easily be implemented, as shown in this video:
MIF Measurements with MAIA
The MAIA allows verification of the MIF values used to calculate the RFail. The MAIA can be operated over the air interface using the built-in ultra-broadband planar log spiral antenna (supported frequency range: 698–6000 MHz) or in conducted mode using the coaxial SMA 50 Ω connector (supported frequency range: 300–6000 MHz).
Figure 1: MIF Measurement of a 5G new radio frequency range 1 frequency division duplex signal
RF Test Setup Verification
A dedicated Notebook has been developed for verifying the RF interference measurement setup. The user can select and optimize the measurement configuration in a list of predefined templates. Peak values are automatically detected for easy comparison with the measurement targets.
Figure 2: Interpolated E-field Distribution of a CD2450V3 Dipole
RF Audio Interface Level Determination
In Module HAC2019, the RFail is assessed using the (preferred) indirect test procedure described in ANSI C63.19-2019. A 50x50 mm region centered on the speaker location is scanned and the root mean square (RMS) electric field values are recorded at each point. Once the scan is completed, the RFail is obtained by adding the MIF value to the average steady RMS field strength averaged over the measurement area.
Figure 3: Interpolated E-field distribution of a wireless device after MIF scaling
Baseband Magnetic T-Coil Signal
The Jupyter Notebook developed for baseband magnetic T-Coil testing guides the users through the measurement steps described in the ANSI C63.19-2019:
Validation of the measurement equipment using a telephone magnetic field simulator (TMFS) or a Helmholtz coil
Determination of the wireless device drive levels for each audio signal
Measurement of the desired magnetic field with a voice-like signal (desired audio band magnetic (ABM) signal)
Measurement of the frequency response at the maximum location of the desired ABM signal
Measurement of the undesired broadband audio magnetic signal (undesired ABM signal)
The T-Coil coupling qualifying strengths, including determination of the primary and secondary groups, as well the magnetic field frequency response are computed by the post-processor and are available for visualization and reporting.
Figure 4: Evaluation of Primary and Secondary Groups according to ANSI C63.19-2019
The software installer will be available on November 15th.
For further information, please contact us at email@example.com or any of our sales channels.
In response to several user requests, SPEAG has added a quick and intuitive measurement setup feature to its cSAR3D software that is optimal for quick & easy measurements as needed in R&D applications or for special analyses. A new remote interface now provides an exceptional level of convenience for QA testing by allowing sites to use their own automated test software and analysis tools, and extending scalability to multi-site environments.
This SEMCAD X release represents a breakthrough in efficiency. Thanks to its new GPU empowered sub-gridding method, simulations that were impossible to solve can now been solved in a few hours. After a year of devising the essential software infrastructure for the most advanced optimizers, SPEAG is proud to announce the release of a preview version of the Sweeper. Exploit its possibilities right away! In addition, we responded to industry leaders and added special tools for the design and analysis of 5G and of other wireless transmitters.
Watch our recent product video demonstrating the latest advancement of our Dielectric Assessment Kit product line. DAK-TL brings automated precise measurements of dielectric permittivity, conductivity and homogeneity of boards, sheets, foils and small samples of tissues or liquids to your bench – giving you results in seconds.
SPEAG is very happy to provide you with an update of the IEC 106 Standard Meetings in Hangzhou, China, together with exciting new data regarding the performance and regulatory acceptance of DASY6 and cSAR3D.
SPEAG will exhibit at EuMW conference (Hall S11, Booth 134) in London, UK to display and demonstrate new products and solutions optimized for the demanding applications in the wireless, automotive, aerospace, and defence industries.
Verkotan has developed a real over-the-air GPS field test environment into the test laboratory. The test (patent pending) has been developed in close collaboration with the Centre for Wireless Communication (CWC) at University of Oulu, the VTT Technical Research Centre of Finland Ltd, and SPEAG. It allows the testing of a device’s total GPS air performance (chipset, mechanics, and antenna) in one test run.
SPEAG is pleased to announce the release of cSARD6 in Q4 2016, the smart SAR testing solution. cSARD6 integrates cSAR3D with DASY6 and adds new features to create a fully automated and powerful SAR compliance system. This is the most advanced SAR measurement system available today, combining speed and versatility, while maintaining full SAR testing capabilities and compatibility to any of the worldwide standards.