SPEAG has released DASY8 Module HAC2019. The new module provides a guided workflow that demonstrates the compliance of any wireless device with the latest Hearing Aid Compatibility standard ANSI C63.19-2019 and the latest Federal Communication Commission requirements. Module HAC2019 is also available for DASY6.
Demonstrating Hearing Aid Compatibility (HAC) is important for warranting the electromagnetic and operational compatibility of hearing aids and wireless devices. In February 2021, the Federal Communication Commission (FCC) announced the adoption of the 2019 edition of ANSI C63.19, the “American National Standard Methods of Measurement Compatibility Between Wireless Communications Devices and Hearing Aids,” which became effective June 4, 2021. The FCC is giving manufacturers a two-year transition period (i.e., until June 4, 2023) to adopt the requirements of the updated ANSI C63.19-2011 standard. It has also extended the volume control deadline to match this transition period. Beginning June 5, 2023, only test reports that comply with ANSI C63.2019 will be accepted.
Following SPEAG’s tradition of staying ahead of requirements, we started early with the development of Module HAC2019 for DASY8/6 users. This is a completely new software solution based on SPEAG’s latest graphical user interface technology that guides users through the compliance workflow by using specific Jupyter Notebooks. The DASY8/6 system is interfaced via a dedicated application programming interface.
During development, we have been meticulous in ensuring that we reuse as much as possible of the previous HAC hardware in order to save resources and costs for our customers. The only element that has been replaced is the audio interference analyzer (AIA), which has been superseded by the modulation and audio interference analyzer (MAIA) to evaluate the modulation interference factor (MIF).
DASY8/6 Module HAC2019 consists of four Jupyter Notebooks that represent the different steps of the compliance workflow:
MIF measurements with MAIA
Simplified verification of the radiofrequency (RF) test setup using RF emission calibration dipoles
Evaluation of the wireless device RF interference potential by determining the RF audio interference level (RFail)
Measurement of the baseband (audio frequency) magnetic T-Coil (Tele-Coil) signal from a wireless device.
The workflow can easily be implemented, as shown in this video:
MIF Measurements with MAIA
The MAIA allows verification of the MIF values used to calculate the RFail. The MAIA can be operated over the air interface using the built-in ultra-broadband planar log spiral antenna (supported frequency range: 698–6000 MHz) or in conducted mode using the coaxial SMA 50 Ω connector (supported frequency range: 300–6000 MHz).
Figure 1: MIF Measurement of a 5G new radio frequency range 1 frequency division duplex signal
RF Test Setup Verification
A dedicated Notebook has been developed for verifying the RF interference measurement setup. The user can select and optimize the measurement configuration in a list of predefined templates. Peak values are automatically detected for easy comparison with the measurement targets.
Figure 2: Interpolated E-field Distribution of a CD2450V3 Dipole
RF Audio Interface Level Determination
In Module HAC2019, the RFail is assessed using the (preferred) indirect test procedure described in ANSI C63.19-2019. A 50x50 mm region centered on the speaker location is scanned and the root mean square (RMS) electric field values are recorded at each point. Once the scan is completed, the RFail is obtained by adding the MIF value to the average steady RMS field strength averaged over the measurement area.
Figure 3: Interpolated E-field distribution of a wireless device after MIF scaling
Baseband Magnetic T-Coil Signal
The Jupyter Notebook developed for baseband magnetic T-Coil testing guides the users through the measurement steps described in the ANSI C63.19-2019:
Validation of the measurement equipment using a telephone magnetic field simulator (TMFS) or a Helmholtz coil
Determination of the wireless device drive levels for each audio signal
Measurement of the desired magnetic field with a voice-like signal (desired audio band magnetic (ABM) signal)
Measurement of the frequency response at the maximum location of the desired ABM signal
Measurement of the undesired broadband audio magnetic signal (undesired ABM signal)
The T-Coil coupling qualifying strengths, including determination of the primary and secondary groups, as well the magnetic field frequency response are computed by the post-processor and are available for visualization and reporting.
Figure 4: Evaluation of Primary and Secondary Groups according to ANSI C63.19-2019
The software installer will be available on November 15th.
For further information, please contact us at firstname.lastname@example.org or any of our sales channels.
Zurich Med Tech, a sister company of SPEAG, has developed iSEG, a medical image segmentation tool specially designed to construct realistic 3D human or animal models for SEMCAD X from medical image data. iSEG, the most versatile medical image segmentation tool currently available, quickly and easily creates highly detailed 3D human or animal models based on CT and MRI images in standard image formats. The resulting models can be used to study electromagnetic exposure and induced heating in living tissue.
The final release of DASY52 is a major upgrade representing the next leap in quality compliance testing of mobile phones and general near-field evaluations. The main new improvements are in the new GUI, Report Generator, Ultra-Fast SAR Procedures, and Advanced Probe Response Linearization. Many more features that make the system compatible with the revised standards are expected to be released by the end of the year.
Schmid & Partner Engineering AG (SPEAG) is now accredited for iSAR calibration to the rigorous ISO/IEC 17025 quality standards. ISO/IEC 17025 is a laboratory quality standard that is widely recognized internationally. The addition of iSAR calibration to the SPEAG quality system provides a large portfolio of products for which SPEAG provides accredited calibration.
Schmid & Partner Engineering AG (SPEAG) is pleased to announce a new partnership with Dellsoft Technologies, who will now act as a reseller of SEMCAD X in India. Dellsoft is a leading technical software company in India. Dellsoft's vision is to achieve international standards of excellence and to be the leading supplier of technical software, learning programs, and other teaching platforms. Many years of experience in the software business and a strong technical support team will facilitate their efforts in providing SEMCAD X to companies in India.