The Whole-Body Back (WBBDuke) phantom shape corresponds to the skin shell of the IT’IS Foundation’s Virtual Population (ViP) model Duke.
The main application is the demonstration of specific absorption rate (SAR) compliance of devices mounted to the back of a person, such as wireless-linked backpacks or charging vests.
This phantom enables easy cross-validation of SAR measurements with simulations performed in Sim4Life.
A cover, which prevents evaporation of the tissue simulating liquid (TSL), is delivered with each unit.
|Standard Compatibility||Measurement procedures described for applications-specific phantoms in IEC/IEEE 62209-1528-U|
|Liquid Compatibility||The phantom shell is compatible with SPEAG's TSL (sugar and oil based). Use of other liquids may render the phantom warranty void (see note or consult SPEAG support).|
2 ± 0.2 mm
Length: 2000 mm
Width: 750 mm
Software: cDASY6 V6.8+
|Additional Documentation||Usage & Validation of Specific Phantoms|